Having same issue. Was there a solution to this?
Make sure you trim off the dotted lines on the top, left and right sides. I struggled with this when we first purchased the ISIS 2, and ever since I started trimming the dashed lines off I haven't had a single issue.
This worked!! Thank you. Cutting the dashed lines off fixes the issue
Which software are you using to scan the charts? You could try a trial version of any other software and see if that works. Make sure that the TC1617 was not scaled up or down. You could try to shift the chart 2 or 3 mm to the right and see if that helps.
Multiple softwares. With the TC1617 there is really no room to move left/right. But cutting the dash lines off the chart solves the issue. Thanks
What software are you using? I had the same problem using Curve 4. I switched to reading the TC 1617 into i1Profiler then exporting the data into Curve 4 and it works great.
1800 Diagonal Road, Suite 320
Alexandria, VA 22314
Code of Conduct
© 2021 Idealliance