That is the paper we are using on an Epson P-7000 with an EFI Fiery RIP. Our other proofers run on ORIS. We are getting the closest results on the Fiery to G7 Colorspace, passing it easily. We are also adding a G7 curve after the linearization but before we make the output profile. We are getting excellent results with that.
Mike Todryk | Color Technical Specialist
IWCO Direct | iwco.com
O: 952-470-5659 | C: 952-797-3940
We do the EFI Fiery calibration with the Spectroproofer on the printer. We read in the P2P for G7 calibration with either an FD-9 or our iO table, usually the FD-9. Same with the TC1617 that we use for substrate profiling. Gives us a wicked match on the G7 grays.
The CIE Lab values I'm getting for the substrate is L: 94.03 a: -0.1 b: -1.01; M0.
We measure a TC1617 weekly on all of our proofers to check for G7 Colorspace compliance. I attached the metrics from our last reading.
1800 Diagonal Road, Suite 320
Alexandria, VA 22314
Code of Conduct
© 2021 Idealliance